- Open Access
Automated longitudinal monitoring of in vivo protein aggregation in neurodegenerative disease C. elegans models
Molecular Neurodegenerationvolume 11, Article number: 17 (2016)
While many biological studies can be performed on cell-based systems, the investigation of molecular pathways related to complex human dysfunctions – e.g. neurodegenerative diseases – often requires long-term studies in animal models. The nematode Caenorhabditis elegans represents one of the best model organisms for many of these tests and, therefore, versatile and automated systems for accurate time-resolved analyses on C. elegans are becoming highly desirable tools in the field.
We describe a new multi-functional platform for C. elegans analytical research, enabling automated worm isolation and culture, reversible worm immobilization and long-term high-resolution imaging, and this under active control of the main culture parameters, including temperature. We employ our platform for in vivo observation of biomolecules and automated analysis of protein aggregation in a C. elegans model for amyotrophic lateral sclerosis (ALS). Our device allows monitoring the growth rate and development of each worm, at single animal resolution, within a matrix of microfluidic chambers. We demonstrate the progression of individual protein aggregates, i.e. mutated human superoxide dismutase 1 - Yellow Fluorescent Protein (SOD1-YFP) fusion proteins in the body wall muscles, for each worm and over several days. Moreover, by combining reversible worm immobilization and on-chip high-resolution imaging, our method allows precisely localizing the expression of biomolecules within the worms’ tissues, as well as monitoring the evolution of single aggregates over consecutive days at the sub-cellular level. We also show the suitability of our system for protein aggregation monitoring in a C. elegans Huntington disease (HD) model, and demonstrate the system’s ability to study long-term doxycycline treatment-linked modification of protein aggregation profiles in the ALS model.
Our microfluidic-based method allows analyzing in vivo the long-term dynamics of protein aggregation phenomena in C. elegans at unprecedented resolution. Pharmacological screenings on neurodegenerative disease C. elegans models may strongly benefit from this method in the near future, because of its full automation and high-throughput potential.
The growing incidence of neurodegenerative diseases (NDs) urges for a complete understanding of the molecular processes underlying neurodegeneration, as a first step towards the final promise of a new class of therapeutics for these diseases. Cellular models have been exploited for some of these studies [1, 2], but the high complexity of the molecular mechanisms implicated in NDs increasingly demands in vivo models for the investigation of complex phenotypes, which are determined by the interplay among different tissues and pathways . The nematode Caenorhabditis elegans represents a very convenient model organism for such in vivo tests, mainly because of its very fast life cycle, combined with the ease of its genetic manipulation and the relatively high level of conserved mechanisms between C. elegans and humans . In the last two decades, several protein-misfolding disorders, including age-related NDs, have been successfully modeled in C. elegans indeed ; libraries of transgenic worms are currently available for the research of the molecular mechanisms underlying Alzheimer’s, Parkinson’s and Huntington’s diseases, as well as ALS . Transgenic expression of disease genes in C. elegans is typically visualized via fluorescently tagged proteins within its transparent tissues. In most of the NDs, specific proteins self-assemble into aggregated species and cellular toxicity can be induced by the protein misfolding and aggregation process itself . Therefore, the spatio-temporal-resolved observation of protein expression and aggregation, associated with the quantification and localization of these aggregates is a key analytical method for the in vivo monitoring of disease evolution. Unfortunately, conventional C. elegans handling and imaging techniques do not allow accurate monitoring of aggregate progression in individual worms over time, since nematodes are typically cultured in large populations on agar plates and irreversibly immobilized by means of anesthetics for high-resolution imaging.
The advent of microfluidics within the C. elegans research community is progressively revolutionizing the field [6–9]. In particular, several miniaturized devices proved their potential in neurobiology studies, such as the investigation of C. elegans oxygen sensation , olfactory  and chemosensory  neuronal activity, exploratory and learning behavior , neurotoxin-induced responses , neuromuscular function , and nerve regeneration [16, 17]. In the neurodegeneration research field, Càceres et al.  recently proposed a microscale system for high-throughput visual screens on worms. This system exploited a curved microchannel geometry to trigger the positioning of nematodes into lateral orientations and facilitate the inspection of D-type motor neurons. Although this device allowed efficiently screening mutants carrying neurodegenerative defects, it did not permit longitudinal monitoring of the worms. Other microfluidic platforms have instead demonstrated the feasibility of continuous worm culture and observation. For example, Krajniak et al.  showed the microfluidic culture of L1-L3 larvae over periods of 12–36 h, whilst introducing a method for worms’ reversible immobilization based on a thermo-sensitive sol–gel transition. Other studies demonstrated the applicability of this immobilization method in different microfluidic formats [20–22]. However, protein aggregation monitoring within ND disease models typically requires worm culture and repeated high-resolution imaging of the same worm over significantly longer time periods (e.g. > 3 days). This imposes severe requirements in terms of system robustness and automation, related to the simultaneous and strict control of environmental conditions, like worm feeding, fluidic exchanges, temperature of the microfluidic environment, etc.. In this perspective, Rohde et al.  demonstrated an elegant automated system for in vivo time-lapse imaging and high-throughput screening of C. elegans in standard multiwell plates, which employed an in-well cooling apparatus for reversible worm immobilization. However, the use of this device for protein aggregation monitoring at single animal resolution is less trivial, as it did not have microfluidics on-board and could not exploit brightfield transmission microscopy as analytical tool. Here we introduce a microfluidic-based methodology for long-term and high-resolution monitoring of protein aggregation and automated analysis of C. elegans ND models. Specifically, we demonstrate the feasibility of in vivo observation, over 4 days at single animal resolution, of SOD1 aggregation in the AM725 C. elegans transgenic strain, which we use as a biological model system for the investigation of the human ALS disease. This is enabled by our microfluidic platform, which co-integrates the following options and functionalities: (i) a method for fast confinement of worms of desired age in microfluidic chambers, by means of pure passive hydrodynamics with no need of any active components, such as integrated valves; (ii) a technique for continuous worm feeding and progeny removal, to preserve the on-chip worm identity over long-term studies; (iii) a method for reversible C. elegans immobilization using a hydrogel, enabling high-resolution imaging at arbitrarily selected moments of their whole lifespan; (iv) an integrated active temperature control system, both to set precise environmental conditions for C. elegans maintenance and to automatically steer the worm immobilization/release process; (v) a compact device assembly, readily adaptable to host different microfluidic designs and suitable for automated multi-dimensional imaging on any upright or inverted microscope.
Results and discussion
Our worm culture and imaging platform features different components (Fig. 1a). Worms are manipulated inside a monolithic polydimethylsiloxane (PDMS) microfluidic chip, conceived as a simple 1-layer device and operating via pure passive hydrodynamics, with no need of any active valving system. The chip is bonded to a standard 150 μm-thick glass coverslip for accurate worm imaging through high-magnification oil immersion objectives. An aluminum frame is specifically designed to host the chip and set a well-defined temperature distribution over its whole three-dimensional (3D) geometry. Both the frame and the top of the PDMS chip are positioned in contact with a thermoelectric module used to set the temperature of the assembly. A central hole in the Peltier module allows light transmission through the PDMS chip, therefore enabling worm imaging via transmission microscopy. A well-dimensioned heat sink ensures the dissipation of excess heat produced by the thermoelectric module; a thermally insulating holder allows positioning the device on the microscope stage while preventing thermal dissipation. The entire structure is held together by screws and springs at the four corners, in order to ensure good thermal contacts throughout the stack. A resistive temperature detector (RTD), in contact with the glass substrate, is employed to sense the temperature of the microfluidic device. The signal measured by the sensor is exploited to set the power provided to the thermoelectric module in a closed-loop configuration, for active control and constant monitoring of the temperature experienced by the worms inside the chip. The feedback loop management is committed to a portable PID controller (Fig. 1b), provided with a software interface. The microfluidic flow inside the device is regulated by computer-controlled syringe pumps. The tubing connecting the chip to the external syringes are directly plugged to the sidewalls of the PDMS device. To get this peculiar configuration, we casted PDMS inside a specifically designed mold, allowing to shape its whole 3D structure (Additional file 1: Supplementary Note 1). Lateral tubing connections are employed to partially embed the tubes in the aluminum thermalization frame, therefore putting them in contact with the Peltier module as well (Fig. 1c). This allows pre-thermalizing the liquids prior to their injection inside the PDMS chip and opens the possibility of tuning the chip temperature through the injected liquid as well, as will be clarified further. Moreover, the lateral positioning of the microfluidic tubing makes our device readily suitable for imaging on both upright and inverted microscopes. Size and shape of the device holder allow perfect fit with any microscope stage or equipment compatible with standard 60 mm Petri dishes. The imaging area available for transmission microscopy has about 15 mm in diameter, corresponding to the central hole of the thermoelectric module (Fig. 1c). This represents the only geometrical constrain in the design of the PDMS chip, thus offering full versatility in using multiple microfluidic layouts. In particular, for our studies, we use three different microfluidic architectures, featuring matrices of worm culture chambers of different shapes and sizes (Fig. 1d).
Worm arraying via passive valves
Passive hydrodynamics allows avoiding the need of additional fluidic control layers or active valving systems, which would complicate use and design of the chip and would reduce its ease of automation. The first critical protocol steps are typically represented by synchronization, loading and distribution of the worms inside the device. In our platform, all these operations are simultaneously accomplished by simply injecting a mixed worm suspension into the microfluidic chip at a proper flow rate, along the In1-Out1 direction (Fig. 2a). We mainly worked with three different chip designs (Fig. 1d) for selecting either L1, L2/L3 or L4 larvae, as these are commonly employed stages for worm synchronization and successive analyses. Size and shape of the microfluidic channels connecting adjacent chambers along the In1-Out1 direction are designed to ensure uniform flow distribution across the chambers and allow access to the chip only to worms which are younger than the desired age – and thus smaller than a certain size. More specifically, we exploit the flexible nature of PDMS to trigger a passive valving mechanism  in the channels (zoom of Fig. 2a). In the “L1 design” (Fig. 1diii), only L1 worms can access the culture chambers because of their smaller size, which is compatible with the passage through microfluidic channels sizing 8 × 14 μm2 in section. When a 1 s pulse of 2 μL inflow is injected into the device, a slight overpressure instantaneously builds up inside the channels, causing a fast temporary increase of their section. This triggers the passage of L1 larvae through the channels (Fig. 2b and Additional file 2: Video S1). The end of the pulse results then in worm confinement inside the chambers, since each channel immediately returns to its initial shape, preventing any spontaneous passage of worms. A few subsequent inflow pulses allow distributing L1 larvae over the whole matrix of 32 chambers in a few seconds. The same principle is employed in the L4 and L2 chip designs (Fig. 1di-dii), for the automated dispensing of larvae in a single culture chamber or in the 4 chamber array. In this case microchannels (60 × 14 μm2 and 30 × 14 μm2 in section for the L4 and L2 designs, respectively) are sized as such to block the passage of adult worms, confine larvae of desired size inside the chambers by the passive valving effect, while directly washing all the smaller larvae out of the chip. Eventually, direct user control and iteration of the worm loading protocol can be used to adjust the worm distribution until a desired number of worms per chamber is obtained.
Temperature control system design and characterization
We investigate the performance of our temperature control system both theoretically (Additional file 1: Supplementary Note 2) and experimentally. The setup can be operated either in “closed-loop configuration” – by means of the PID controller – or in “open-loop mode”, i.e. by providing the Peltier module with a constant electrical power. The former configuration results in fully automated control of the setup temperature, the latter option has the advantage of allowing device operation with no need of a feedback sensing system. To extract the “open loop” calibration curve of our device, we first characterize its response for different constant cooling electrical powers (Fig. 3a). In this study, the temperature is measured by means of a 1.6 × 2.5 mm2 Pt1000 sensor positioned on the bottom face of the glass coverslip, aligned with the center of the microfluidic chip (at x = y = 0 in the insert of Fig. 3b). Because of the reduced thickness of the glass substrate, negligible temperature variations are expected between top and bottom face of the coverslip. As expected, for different cooling power in the “open loop” configuration, the system is cooled down to specific steady-state temperatures, set by the equilibrium between thermoelectric cooling and heat convection in the surrounding air. Starting from an ambient temperature of 24 °C, stable temperatures down to 10 °C can be reached at the chip center, with a clear linear dependence on the electrical power that is applied to the Peltier module (Fig. 3b). Because of the specific geometry of the device, the spatial temperature distribution is not constant throughout the chip area, as observed via measurements at different locations from the coverslip center (Fig. 3c). By fitting these data with the corresponding simulated results, we empirically extract the heat transfer coefficient h, which models heat convection in our system (Equation S2). A slight deviation in the fit between measured and simulated data is observed only close to the PDMS chip edge (y = ± 10 mm) and attributed to imperfections of the contact surfaces between the chip and the metallic frame. We moreover simulate the spatial temperature distribution for different cooling powers, showing again good agreement between theoretical and experimental results (Fig. 3c). The dynamics of heat exchange in our system can be efficiently studied by normalizing the curves of Fig. 3a with respect to the external temperature T ext and the steady-state temperature T eq for each applied power. Normalized data prove to be independent from the electrical power and allow defining the calibration curve that describes the temperature evolution of the device in the “open loop” configuration (Additional file 1: Supplementary Note 3). Further enhancement and fine-tuning of the cooling efficiency of the device are moreover enabled by controlling the microfluidic inflow speed, which allows accurately modulating the liquid pre-thermalization (Additional file 1: Supplementary Note 4).
Automated worm culture and imaging protocol
The cross-shape of our chips, with in- and outflows along two orthogonal directions, is designed to decouple the worms’ dispensing operation (in the In1-Out1 direction) from the worm culture and imaging protocol. Along the In2-Out2 direction, adjacent chambers are connected by narrow filters – 5 × 14 μm2 in section – allowing perfusion of liquids across the whole chamber matrix, while preventing any inter-chamber exchange of worms, even under over-pressure conditions. Also, after worm dispensing, the In1-Out1 flow direction can be employed during the culture experiments for evacuating the progeny of the adult worms under analysis. Each switching between the two flow directions is simply controlled by two external valves at the two chip outlets.
We develop a fully automated protocol for worm culture and high-resolution imaging inside the device (Fig. 4a). Worms are cultured on-chip at a desired temperature (typically in the 20–25 °C range) by operating the temperature controller in a closed-loop configuration. Escherichia coli suspension is perfused at desired rate along the In2-Out2 direction for worm feeding (Fig. 4ai). For screening purposes and high-throughput-like experiments, our platform can be readily used in imaging experiments with standard low-magnification microscope objectives. However, for ultimate imaging accuracy and repeatable observation of the worms through high-magnification objectives, C. elegans need to be temporarily immobilized in a reversible manner. To fulfil this requirement, we employ the thermoreversible gelation of a PF127 solution around the worms, as a minimally invasive C. elegans immobilization technique. Previously reported results, in fact, proved that both thermocycling and the exposure to PF127 for repeated imaging cycles do not alter viability, development and physiology of C. elegans . Prior to imaging, the chip temperature is set at 15 °C and a liquid solution of 25 % w/v PF127 is injected into the culture chambers (Fig. 4aii). The chip temperature is then raised to 25 °C, to trigger the gelation of the Pluronic solution. This significantly increases the viscosity ensuring stable worm immobilization (Fig. 4aiii and Additional file 3: Video S2). Upon imaging, the chip temperature is brought back to 15 °C, to release the worms and wash the PF127 solution out of the chambers, by replacing it with E. coli suspension to restart worm culture and feeding (Fig. 4aiv). The whole protocol can be iterated many times per experiment and strongly relies on accurate temperature control, especially to trigger the worm immobilization/release process. The closed-loop temperature management system plays therefore a crucial role for the device automation. The spatial distribution of temperature over the whole chip geometry is carefully considered, to minimize inter-chamber variations and expose all the worms to the same experimental conditions. We demonstrate via FEM simulations that our device ensures temperature uniformity within 1 °C difference, even when using the largest chamber matrix, and this both during worm culture (at 20 °C) and PF127 injection and washing (at 15 °C) (Fig. 4b). PID parameters are chosen to minimize any overshoots outside of the standard temperature range for worm culture (15–25 °C) (Fig. 5a). Active temperature control allows moreover fast and accurate steering of the sol–gel transition of the PF127 solution. In our device, such a transition occurs about 1 min after the temperature setpoint shift from 15 °C to 25 °C , i.e. worms are ready for high-resolution imaging in about 2–3 min, when the viscosity of the PF127 gel reaches its highest value and guarantees worm immobilization (Fig. 5b). We experimentally determine the viscosity at different temperatures by dispensing a ~2 mL PF127 solution over the bottom plate of a cone-plate viscometer (Bohlin Gemini Malvern, UK) and measuring using a shear rate of 10 s−1. This value is chosen to be comparable with typical shear rates associated to the C. elegans swimming motion (1–20 s−1) . Finally, we attribute a crucial role to the liquid inflow pre-thermalization as far as the PF127 injection process is concerned. Even in liquid phase, the PF127 solution behaves in fact as a very viscous non-newtonian fluid, and, as such, it is challenging to manipulate through syringes, tubes and microchannels. Pre-thermalizing the PF127 inflow in the metallic frame to temperatures even lower than 15 °C allows further reducing the solution viscosity at the chip inlet and improving the control over its injection.
Long-term protein aggregation analysis in an amyotrophic lateral sclerosis (ALS) C. elegans model
Upon full validation of worm viability and culture in our device (Additional file 1: Supplementary Note 5), we employ it to monitor the dynamics of protein aggregation in an ALS disease C. elegans model. ALS is a neurodegenerative human disease causing selective death of motor neurons. In Europe and the United States, this disease affects about 2 people per 100,000 per year , with average survival from onset to death of only 3–5 years and no cure currently available . In ALS, such as in many other neurodegenerative diseases, cellular toxicity could be mediated by the misfolding and aggregation of specific mutant proteins [28–30]. Time-resolved imaging and quantification of these aggregates is hence a key phenotyping method to monitor disease progression. In our study, we employ an ALS C. elegans model expressing mutated human SOD1-YFP fusion proteins in the body wall muscle cells (AM725 transgenic worms)  While this model presents some limitations for investigating neurodegeneration per se, compelling evidence in literature already proved that C. elegans models expressing protein aggregation in muscle cells are useful for screening neurodegenerative phenotypes about Parkinson’s and Huntington’s diseases [32–34]. In our device, worms are loaded at the L2 stage into a 4-chamber microfluidic chip (Fig. 1dii), where their distribution is adjusted to isolate a single worm in each chamber. Automated on-chip culture of these worms is then conducted according to the previously described protocols. Correct worm feeding, development and reproduction are constantly monitored via brightfield time-lapse imaging (Fig. 6a). Automated image processing algorithms on these time-lapse pictures allow moreover extracting detailed information about the worms’ growth rate, not only at a population level, but also at single-worm resolution (Fig. 6b-c). Specifically, the area occupied by each worm is here used as a quantitative indicator of growth. Data about the growth rate of individual worms allows preserving detailed information on the development of each nematode (Fig. 6b), while the average growth trend is found to follow a well-defined sigmoidal dependence on time (Fig. 6c). At desired moments during worms’ development (e.g., in our study, 43 h, 60 h and 91 h upon worm injection), the reversible worm immobilization protocol for high-resolution imaging (see Fig. 4a) is employed. At each observation and in a matter of a few minutes, all the worms are perfectly immobilized in a PF127 gel matrix and SOD1-YFP expression can be monitored within their tissues via fluorescent microscopy through high-magnification objectives. A first analysis at 10× magnification allows observing protein aggregates within the whole body of each worm (Fig. 7a). Aggregation patterns observed in our system are in line with previously reported results: while wild-type SOD1 is known to exhibit only diffuse fluorescence in body wall muscle cells, mutant SOD1 proteins present punctuated fluorescent patterns . Interestingly, we find that the temporal evolution of SOD1-YFP aggregation features some observable worm-to-worm variability (Fig. 7b), and follows an increasing trend in the considered temporal window (43–91 h after on-chip loading at the L2 stage) (Fig. 7c).
A second set of studies is then conducted by imaging the immobilized worms through a 63× oil immersion objective (NA 1.4). Many aggregates in AM725 worms, unlike in other analogous SOD1-transgenic strains – e.g. pUnc-54::SOD1-G85R::YFP (G85R) and pUnc-54::SOD1- G93A::YFP (G93A) –, appear as irregular, elongated foci . This feature could be observed before by confocal imaging of paralyzed worms. We are now able to confirm this observation in alive immobilized worms and provide a precise sub-cellular mapping of their protein aggregation pattern at high spatio-temporal resolution by using a standard fluorescent microscope (Fig. 7d). Moreover, the possibility to take quasi-instantaneous brightfield and fluorescent pictures in our chip allows accurately locating each fluorescent signal inside the C. elegans body. In combination with reversible worm immobilization, this opens the possibility of following the temporal evolution of protein aggregation at precise locations within the worm tissues and monitoring aggregate progression in vivo, not only at single-worm, but even at the single-cell level (Fig. 7e).
Long-term protein aggregation analysis in a Huntington disease (HD) C. elegans model
To ascertain that this approach is not specific only for the ALS model, we monitor the dynamics of protein aggregation in a different C. elegans model of neurodegenerative diseases, i.e. a Huntington disease (HD) model (Additional file 1: Supplementary Note 6). More particularly, we employ a HD model expressing YFP fused to stretches of 35 glutamine residues (AM140 transgenic strain) . Our system allows the observation of distinct aggregation patterns for this model and allows following the long-term evolution of both aggregate size and number, with results in line with what has been previously reported . Furthermore, the specificity of the approach to monitor the aggregates is validated by using a strain expressing the YFP only in the body wall muscles (AM134 strain) and showing only a diffuse fluorescence in these tissues .
Modifying long-term protein aggregation in an ALS C. elegans model by doxycycline treatment
ALS is the most common motor neuron disease in adults, causing the selective loss of the spinal and cranial motor neurons cells that directly connect the brain to muscles. The disease is characterized by rapidly progressive paralysis and death from respiratory failure, typically within 2–3 years of symptom onset . There are currently no effective cures for ALS, although Riluzole was found to slow the rate of progression and prolongs survival by 3 months . Mitochondrial accumulation of misfolded mutant SOD1 has been proposed as one possible trigger of motor neuron death . Mitochondrial degeneration , vacuolization and swelling  are pathological features of both familial human ALS cases and mutant SOD1 mouse models. Hence, damaged or dysfunctional mitochondria are common in most familial ALS cases and could represent a potential therapeutic target for treatment of this disease. Activation of the mitochondrial unfolded protein response (UPRmt) recently emerged as an interesting approach to restore a pool of healthy, functional mitochondria in stressed animals and maintain organism health . Indeed, genetic or pharmacological inductions of UPRmt can extend the lifespan in various model organisms, including worms [41–43]. Beneficial effects of UPRmt rely on the massive expression of mitochondrial chaperones that help to repair the damages caused in the organelle . Doxycycline –an antibiotic belonging to the tetracycline family– proved to be beneficial for certain physiological aspects, like for increasing the motility in worms and flies, and extending the lifespan of worms through the activation of UPRmt [42, 45]. To determine whether UPRmt could prevent the ALS progression in C. elegans, we treat worms with doxycycline and monitor the aggregate formation. First, in order to evaluate the efficacy of doxycycline treatment in our microfluidic system, we need to monitor the induction of the UPRmt. We therefore use a transgenic strain of worms that reports on the activity of the UPRmt with integrated GFP genes driven by the regulatory DNA region of the mitochondrial chaperone hsp-6 . In these transgenic worms, an increase of hsp-6::gfp expression is indicative of the presence of a mitochondrial stress and the subsequent induction of the UPRmt . As previously observed on solid nematode growth media (NGM) plates , doxycycline dose-dependently also induces the expression of the hsp-6::gfp on-chip, revealing the activation of the UPRmt (Fig. 8a). At the same time, doxycycline also induces a significant growth delay, which is another known physiological effect produced by the antibiotic (Fig. 8b) . Based on these first observations obtained from the hsp-6::gfp expressing worms, we decide to use a concentration of 15 μg/mL doxycycline in our microfluidic experiments on the ALS worm model, as (1) this concentration is comparable to the concentrations showing a significant effect on the worm lifespan and on the UPRmt on solid plates, and (2) the impact on the growth of this concentration is minor. Moreover, we treat ALS worms with doxycycline and compare the aggregation pattern with that of an untreated population. Strikingly, doxycycline treatment suppresses almost completely the size expansion of the aggregates, without affecting their number (Fig. 8c-d).
Protein aggregation diseases are often associated with movement disorders, and worm models of neurodegenerative diseases also recapitulate this feature [31, 35, 47–49]. Interestingly, doxycycline treatment significantly slows down the progressive loss of motility in the ALS strain, as observed on NGM plates (Fig. 8e). Since doxycycline is known to affect the mitochondrial translation in eukaryote systems , we then employ a genetic approach to impact on the mitochondrial translation machinery. We silence by RNAi the gene mrps-5 (mitochondrial ribosomal protein S5), an important molecular actor for the mitochondrial translation , and we could observe similar effects on the motility in the ALS model (Fig. 8f). All together, these results show that doxycycline can prevent the size expansion of aggregates in a C. elegans model of ALS, and this improvement correlates with a delay in the loss of motility.
We introduce a new analytical technique and device for automated time-resolved studies on C. elegans nematodes down to single-worm resolution. Our platform is based on a multi-functional approach, where several functionalities are integrated into a single miniaturized device, to allow fully automated worm analyses. The device is moreover designed to be compatible with different microfluidic designs and readily suitable for different sets of studies.
In our platform, C. elegans nematodes are loaded into a microfluidic chip, where they are directly distributed among a set of culture chambers via a “passive valving” method. Geometrical constraints on the chip allow retaining only worms of desired size inside the device, whereas an automated on-chip culture protocol is established to ensure their correct feeding and development. Active control of the chip temperature ensures moreover running worm cultures at desired temperatures, with minimal variation throughout long-term analyses. For screening purposes and high-throughput-like experiments, our platform can be readily used in imaging experiments with standard low-magnification microscope objectives. Moreover, to allow longitudinal high-resolution imaging of the worms, we optimize an automated procedure for reversible worm immobilization on-chip. This protocol is based on the thermoreversible gelation of PF127 polymer inside the device, managed by the closed-loop temperature control system as well. Any worms’ progeny is periodically washed out of the chip, with no risk of mixing the identities of the worms under analysis. Therefore, tests at single-worm resolution can be easily performed on our platform. Finally, all the microfluidic designs used in our studies are conceived in a “chamber-matrix” format, which allows easy automation of the imaging process as well.
We fully characterize the different functionalities of our platform, both theoretically and experimentally. We demonstrate fast and precise temperature management on the device and provide calibration curves for its use both in open-loop mode and in closed-loop configuration. We characterize the different integrated worm handling protocols – i.e. on-chip worm loading, feeding, immobilization, imaging – and provide details for their use on the platform. We then employ our device to tackle the challenging task of analyzing the dynamics of protein aggregation in ALS worm models over long-term experiments. Our results show that the device ensures reliable culture and reproducible growth rate of the worms over several days. The possibility of isolating single worms in separated chambers allows collecting population statistics, while preserving at the same time all the information related to the single nematodes under test. For high-resolution imaging experiments, we employ the on-chip immobilization protocol to temporarily immobilize the worms in a reversible way and periodically collect data about protein aggregation in their tissues via high-resolution fluorescent imaging. Our results show that the amount of SOD1-YFP aggregates in an ALS C. elegans model (AM725 transgenic worms) linearly increases over the whole analyzed period (i.e. day 1 to day 4 of adult life). Combined brightfield and fluorescent imaging at high magnification allows moreover mapping the geometry of the aggregates, precisely locate them within the tissues of each worm and following their progression over consecutive days. We also demonstrate the suitability of our system for protein aggregation monitoring in a C. elegans Huntington disease (HD) model, and demonstrate the systems’s ability to study long-term doxycycline treatment-linked modification of protein aggregation profiles in the ALS model. In fact, the relatively short period needed for the quantification of a significant increase in SOD1-YFP aggregates opens the possibility for future studies of rapid identification of ALS modifiers . Because of its good performance in terms of automation and versatility, we envision that our system could be employed to address many other challenging biological questions on C. elegans, related in particular to the study of neurodegenerative diseases – such as Parkinson’s, and Alzheimer’s disease – which are all modelled in worms .
Our platform could moreover be used for studies of C. elegans movement disorders, for quantifying other phenotypes such as pharyngeal pumping rates, motility, etc. or more generally could be used in chemical or biological laboratories that do in vivo studies and analyses of multicellular organisms.
Chemicals and materials
Four-inch 550 μm thick Si and float glass wafers, de-ionized water (DIW) were obtained from the Center of Micro- and Nanotechnology of EPFL. GM 1070 SU-8 negative photoresist was purchased from Gersteltec (Pully, Switzerland). PDMS Sylgard 184 was acquired from Dow Corning (Wiesbaden, Germany). 1 mL borosilicate H-TLL-PE syringes were purchased from Innovative Laborsysteme GmbH (Stutzerbach, Germany). Microline ethyl vinyl acetate tube with 0.51 mm inner and 1.52 mm outer diameters was bought from Fisher Scientific (Wohlen, Switzerland). Pluronic F-127 was purchased from Sigma-Aldrich (Buchs, Switzerland). M9 buffer was obtained by adding 3 g KH2PO4, 6 g Na2HPO4, 5 g NaCl, 1 mL 1 M MgSO4, H2O to 1 l and sterilization by autoclaving. S-medium buffer was obtained by adding 10 mL 1 M potassium citrate pH 6, 10 mL trace metals solution (1.86 g disodium EDTA, 0.69 g FeSO4 · 7H2O, 0.2 g MnCl2 · 4H2O, 0.29 g ZnSO4 · 7H2O and 0.025 g CuSO4 · 5H2O, H2O to 1 l) 3 mL 1 M CaCl2, 3 mL 1 M MgSO4, 1 mL [50 mg/mL] carbenicillin, 0.5 mL tween 20 to 1 l S Basal (5.85 g NaCl, 1 g K2HPO4, 6 g KH2PO4, 1 mL [5 mg/mL] cholesterol, H2O to 1 l) and sterilization by autoclaving. Pluronic F127 solution was prepared by diluting 25 % (weight/volume) Pluronic F127 in water. Aluminum and polymethylmetacrylate (PMMA) assembly parts were fabricated at the engineering workshop of EPFL. Thermoelectric modules were bought from TE Technology, Inc. (Traverse City, MI, USA), heat sinks from Advanced Thermal Solutions, Inc. (Norwood, MA, USA) and RTD sensors from Innovative Sensor Technology AG (Ebnat-Kappel, Switzerland), while a proportional-integral-derivative (PID) temperature controller was purchased from BelektroniG GmbH (Freital, Germany).
C. elegans strains and culture
C. elegans strains were cultured at 20 °C on NGM agar plates seeded with the Escherichia coli strain OP50. Strains used were wild-type Bristol N2, AM725 (rmIs290[unc-54p::Hsa-sod-1(127X)::YFP]), AM134 (rmIs126[unc-54p::Q0::YFP]), AM140 (rmIs132[unc-54p::Q35::YFP]) and SJ4100 (zcIs13[hsp-6::GFP]) and were provided by the Caenorhabditis Genetics Center (University of Minnesota). Worms were suspended in S-medium solution prior to each microfluidic experiment. For microfluidic experiments, the E. coli strain HT115 was suspended in S-medium at a concentration of 1.4 × 109 cells/mL. Bacterial feeding RNAi experiments were carried out as described . The clone used was mrps-5 (E02A10.1). For on-plate motility assays, doxycycline was added at the indicated concentration just before pouring the plates. Animals were exposed to compounds from eggs until the day of the experiment.
C. elegans movement was recorded for 45 s at different days of adulthood using a Nikon DS-L2/DS-Fi1 camera and controller setup, attached to both a computer and a standard brightfield microscope. Five plates of worms, with 10 worms per plate were measured in each condition. Using these video recordings, the movement traces of worms during all recording periods were calculated by following the organism centroids using a modified version of the Parallel Worm Tracker for MATLAB . The average worm speed during the recording periods was then calculated for each plate and each condition.
Fabrication of the microfluidic chips
Microfluidic devices were prepared by soft lithography  using 2-layer SU-8 molds. Briefly, conventional photolithography was used to pattern a 14 μm-thick layer of SU-8 photoresist on 4-in. wafers. A ~110 μm-thick layer of SU-8 was then patterned on top of the first one. Layer thicknesses were confirmed by mechanical profilometer measurements. The silicon mold was then diced in 20 mm × 20 mm microchips, which were inserted at the bottom of an aluminum/PMMA mold for PDMS casting (Additional file 1: Supplementary Note 1). 1.5 mm diameter steel pins were used to define the lateral connections of the device for the external tubing insertion. A liquid PDMS mixture (10:1 base:cross-linker weight ratio) was degassed, injected into the mold and cured at 100 °C for 1 h. Upon extraction from the mold, each PDMS chip was bonded by plasma-activation to a 150 μm-thick, 32 × 24 mm2 glass coverslip. The chip was then connected to external tubing and enclosed in the device assembly as reported in Fig. 1a.
Image acquisition and processing
The microfluidic platform was placed within an inverted microscope (Axio Observer, Zeiss) equipped with two illumination systems: (i) a precisExcite High-Power LED illumination system (Visitron, Puchheim, Germany) for brightfield imaging and (ii) a Lambda DG4 illumination system (Sutter instruments, Novato, CA, USA) for fluorescence imaging. The microscope had a motorized xy-stage and the automated imaging process was controlled using VisiView Premier Image acquisition software (Visitron, Puchheim, Germany). Images were acquired through a Hamamatsu Orca-ER CCD camera (Hamamatsu, Solothurn, Switzerland). Image processing was performed with Fiji software (http://imagej.nih.gov/ij; version 1.47b). In particular, worm areas were measured by processing time-lapse brightfield pictures as follows. Each frame was first converted to a binary image by applying a threshold to the full stack of time-lapse images and transforming it into a set of binary masks. Each stack of masks was then analyzed using the “particle analysis” Fiji plugin, which allows directly extracting area values for each picture in the stack. The same method was then applied to the stacks of fluorescence images, in order to calculate aggregate area values. In this case, we applied a systematic thresholding algorithm which was based on a variational approach, assuring that all aggregates in an image were effectively counted (i.e. by not setting the threshold too low), while not resulting in artificial size reduction of the aggregates (which would be the case by fixing a too high threshold). The “particle analysis” plugin allowed measuring the number and the average size of the aggregates identified in each picture.
Additional material is available: supplementary notes on (i) 3D PDMS chip casting with lateral fluidic connections; (ii) temperature control system: theoretical considerations; (iii) heat exchange dynamics in the device; (iv) inflow pre-thermalization study; (v) worm viability and culture tests; (vi) protein aggregation analysis in a Huntington disease C. elegans model. Supplementary videos on: (i) L1 worm loading via passive valves, and (ii) AM725 worm immobilization by PF127 gelation.
Kunze A, Lengacher S, Dirren E, Aebischer P, Magistretti PJ, Renaud P. Astrocyte-neuron co-culture on microchips based on the model of SOD mutation to mimic ALS. Integr Biol. 2013;5:964–75.
Choi SH, Kim YH, Hebisch M, Sliwinski C, Lee S, D'Avanzo C, et al. A three-dimensional human neural cell culture model of Alzheimer's disease. Nature. 2014;515:274–U293.
Li J, Le WD. Modeling neurodegenerative diseases in Caenorhabditis elegans. Exp Neurol. 2013;250:94–103.
Lai CH, Chou CY, Ch'ang LY, Liu CS, Lin WC. Identification of novel human genes evolutionarily conserved in Caenorhabditis elegans by comparative proteomics. Genome Res. 2000;10:703–13.
Nussbaum-Krammer CI, Morimoto RI. Caenorhabditis elegans as a model system for studying non-cell-autonomous mechanisms in protein-misfolding diseases. Dis Model Mech. 2014;7:31–9.
Chronis N. Worm chips: Microtools for C. elegans biology. Lab Chip. 2010;10:432–7.
Hulme SE, Whitesides GM. Chemistry and the Worm: Caenorhabditis elegans as a Platform for Integrating Chemical and Biological Research. Angew Chem Int Edit. 2011;50:4774–807.
Sivagnanam V, Gijs MA. Exploring living multicellular organisms, organs, and tissues using microfluidic systems. Chem Rev. 2013;113:3214–47.
Bakhtina NA, Korvink JG. Microfluidic laboratories for C. elegans enhance fundamental studies in biology. RSC Adv. 2014;4:4691–709.
Zimmer M, Gray JM, Pokala N, Chang AJ, Karow DS, Marletta MA, et al. Neurons Detect Increases and Decreases in Oxygen Levels Using Distinct Guanylate Cyclases. Neuron. 2009;61:865–79.
Chalasani SH, Chronis N, Tsunozaki M, Gray JM, Ramot D, Goodman MB, et al. Dissecting a circuit for olfactory behaviour in Caenorhabditis elegans. Nature. 2007;450. 63.
Chung KH, Crane MM, Lu H. Automated on-chip rapid microscopy, phenotyping and sorting of C. elegans. Nat Methods. 2008;5:637–43.
Qin JH, Wheeler AR. Maze exploration and learning in C-elegans. Lab Chip. 2007;7:186–92.
Ma H, Jiang L, Shi WW, Qin JH, Lin BC. A programmable microvalve-based microfluidic array for characterization of neurotoxin-induced responses of individual C. elegans. Biomicrofluidics. 2009;3(4):44114.
Lockery SR, Hulme SE, Roberts WM, Robinson KJ, Laromaine A, Lindsay TH, et al. A microfluidic device for whole-animal drug screening using electrophysiological measures in the nematode C. elegans. Lab Chip. 2012;12:2211–20.
Guo SX, Bourgeois F, Chokshi T, Durr NJ, Hilliard MA, Chronis N, et al. Femtosecond laser nanoaxotomy lab-on-achip for in vivo nerve regeneration studies. Nat Methods. 2008;5:531–3.
Samara C, Rohde CB, Gilleland CL, Norton S, Haggarty SJ, Yanik MF. Large-scale in vivo femtosecond laser neurosurgery screen reveals small-molecule enhancer of regeneration. Proc Natl Acad Sci U S A. 2010;107:18342–7.
Caceres ID, Valmas N, Hilliard MA, Lu H. Laterally Orienting C. elegans Using Geometry at Microscale for High-Throughput Visual Screens in Neurodegeneration and Neuronal Development Studies. PloS one. 2012;7:e35037.
Krajniak J, Lu H. Long-term high-resolution imaging and culture of C. elegans in chip-gel hybrid microfluidic device for developmental studies. Lab Chip. 2010;10:1862–8.
Krajniak J, Hao Y, Mak HY, Lu H. CLIP-continuous live imaging platform for direct observation of C. elegans physiological processes. Lab Chip. 2013;13:2963–71.
Hwang H, Krajniak J, Matsunaga Y, Benian GM, Lu H. On-demand optical immobilization of Caenorhabditis elegans for high-resolution imaging and microinjection. Lab Chip. 2014;14:3498–501.
Aubry G, Zhan M, Lu H. Hydrogel-droplet microfluidic platform for high-resolution imaging and sorting of early larval Caenorhabditis elegans. Lab Chip. 2015;15:1424–31.
Rohde CB, Yanik MF. Subcellular in vivo time-lapse imaging and optical manipulation of Caenorhabditis elegans in standard multiwell plates. Nat Commun. 2011;2:271.
Chung K, Zhan M, Srinivasan J, Sternberg PW, Gong E, Schroeder FC, et al. Microfluidic chamber arrays for whole-organism behavior-based chemical screening. Lab Chip. 2011;11:3689–97.
Shen XN, Arratia PE. Undulatory Swimming in Viscoelastic Fluids. Phys Rev Lett. 2011;106:208101.
Kiernan MC, Vucic S, Cheah BC, Turner MR, Eisen A, Hardiman O, et al. Amyotrophic lateral sclerosis. Lancet. 2011;377:942–55.
Kiernan MC. ALS and neuromuscular disease: in search of the Holy Grail. Lancet Neurol. 2014;13:13–4.
Jonsson PA, Ernhill K, Andersen PM, Bergemalm D, Brannstrom T, Gredal O, et al. Minute quantities of misfolded mutant superoxide dismutase-1 cause amyotrophic lateral sclerosis. Brain. 2004;127:73–88.
Rakhit R, Robertson J, Vande Velde C, Horne P, Ruth DM, Griffin J, et al. An immunological epitope selective for pathological monomer-misfolded SOD1 in ALS. Nat Med. 2007;13:754–9.
Pratt AJ, Shin DS, Merz GE, Rambo RP, Lancaster WA, Dyer KN, et al. Aggregation propensities of superoxide dismutase G93 hotspot mutants mirror ALS clinical phenotypes. Proc Natl Acad Sci U S A. 2014;111:E4568–76.
Gidalevitz T, Krupinski T, Garcia S, Morimoto RI. Destabilizing protein polymorphisms in the genetic background direct phenotypic expression of mutant SOD1 toxicity. PLoS Genet. 2009;5:e1000399.
Jadiya P, Fatima S, Baghel T, Mir SS, Nazir A. A Systematic RNAi screen of neuroprotective genes identifies novel modulators of alpha-synuclein-associated effects in transgenic caenorhabditis elegans. mol neurobiol. 2015.
Nollen EA, Garcia SM, van Haaften G, Kim S, Chavez A, Morimoto RI, et al. Genome-wide RNA interference screen identifies previously undescribed regulators of polyglutamine aggregation. Proc Natl Acad Sci U S A. 2004;101:6403–8.
van Ham TJ, Thijssen KL, Breitling R, Hofstra RM, Plasterk RH, Nollen EA. C elegans model identifies genetic modifiers of alpha-synuclein inclusion formation during aging. PLoS genetics. 2008;4:e1000027.
Morley JF, Brignull HR, Weyers JJ, Morimoto RI. The threshold for polyglutamine-expansion protein aggregation and cellular toxicity is dynamic and influenced by aging in Caenorhabditis elegans. Proc Natl Acad Sci U S A. 2002;99:10417–22.
Renton AE, Chio A, Traynor BJ. State of play in amyotrophic lateral sclerosis genetics. Nat Neurosci. 2014;17:17–23.
Cozzolino M, Ferri A, Carri MT. Amyotrophic lateral sclerosis: from current developments in the laboratory to clinical implications. Antioxid Redox Signal. 2008;10:405–43.
Wong PC, Pardo CA, Borchelt DR, Lee MK, Copeland NG, Jenkins NA, et al. An adverse property of a familial ALS-linked SOD1 mutation causes motor neuron disease characterized by vacuolar degeneration of mitochondria. Neuron. 1995;14:1105–16.
Kong J, Xu Z. Massive mitochondrial degeneration in motor neurons triggers the onset of amyotrophic lateral sclerosis in mice expressing a mutant SOD1. J Neurosci. 1998;18:3241–50.
Andreux PA, Houtkooper RH, Auwerx J. Pharmacological approaches to restore mitochondrial function. Nat Rev Drug Discov. 2013;12:465–83.
Mouchiroud L, Houtkooper RH, Moullan N, Katsyuba E, Ryu D, Canto C, et al. The NAD(+)/Sirtuin Pathway Modulates Longevity through Activation of Mitochondrial UPR and FOXO Signaling. Cell. 2013;154:430–41.
Houtkooper RH, Mouchiroud L, Ryu D, Moullan N, Katsyuba E, Knott G, et al. Mitonuclear protein imbalance as a conserved longevity mechanism. Nature. 2013;497:451–7.
Durieux J, Wolff S, Dillin A. The cell-non-autonomous nature of electron transport chain-mediated longevity. Cell. 2011;144:79–91.
Jovaisaite V, Mouchiroud L, Auwerx J. The mitochondrial unfolded protein response, a conserved stress response pathway with implications in health and disease. J Exp Biol. 2014;217:137–43.
Moullan N, Mouchiroud L, Wang X, Ryu D, Williams EG, Mottis A, et al. Tetracyclines Disturb Mitochondrial Function across Eukaryotic Models: A Call for Caution in Biomedical Research. Cell reports. 2015. doi:10.1016/j.celrep.2015.02.034.
Yoneda T, Benedetti C, Urano F, Clark SG, Harding HP, Ron D. Compartment-specific perturbation of protein handling activates genes encoding mitochondrial chaperones. J Cell Sci. 2004;117:4055–66.
Brignull HR, Moore FE, Tang SJ, Morimoto RI. Polyglutamine proteins at the pathogenic threshold display neuron-specific aggregation in a pan-neuronal Caenorhabditis elegans model. J Neurosci. 2006;26:7597–606.
Lakso M, Vartiainen S, Moilanen AM, Sirvio J, Thomas JH, Nass R, et al. Dopaminergic neuronal loss and motor deficits in Caenorhabditis elegans overexpressing human alpha-synuclein. J Neurochem. 2003;86:165–72.
Regitz C, Wenzel U. Amyloid-beta (Abeta1-42)-induced paralysis in Caenorhabditis elegans is reduced by restricted cholesterol supply. Neurosci Lett. 2014;576:93–6.
Riboldi G, Nizzardo M, Simone C, Falcone M, Bresolin N, Comi GP, et al. ALS genetic modifiers that increase survival of SOD1 mice and are suitable for therapeutic development. Prog Neurobiol. 2011;95:133–48.
Kamath RS, Martinez-Campos M, Zipperlen P, Fraser AG, Ahringer J. Effectiveness of specific RNA-mediated interference through ingested double-stranded RNA in Caenorhabditis elegans. Genome Biol. 2001;2:RESEARCH0002.
Ramot D, Johnson BE, Berry Jr TL, Carnell L, Goodman MB. The Parallel Worm Tracker: a platform for measuring average speed and drug-induced paralysis in nematodes. PLoS One. 2008;3:e2208.
Xia YN, Whitesides GM. Soft lithography. Angew Chem Int Ed. 1998;37:551–75.
Work in MG laboratory was supported by the Ecole Polytechnique Fédérale de Lausanne and the EU Ideas program (ERC-2012-AdG-320404). The authors thank R. Padovani, A. Sayah (Laboratory of Microsystems LMIS2 at EPFL), and P. Maoddi (Physics Department of the European Organization for Nuclear Research - CERN) for very fruitful discussions about the project and P. Abdel-Sayed (Biomedical Orthopedics Laboratory at EPFL) for his help with PF127 viscosity measurements. JA is the Nestlé Chair in Energy Metabolism. Work in the JA laboratory is supported by the Ecole Polytechnique Fédérale de Lausanne, the EU Ideas program (ERC-2008-AdG-23118), the NIH (R01AG043930), the Swiss National Science Foundation (31003A-124713) and Systems X (51RTP0-151019). LM is supported by a FRM fellowship.
The authors declare that they have no competing interests.
MC designed the platform and developed the method; GK and MC characterized the platform, optimized the methodology and performed the experiments; LM and GK provided the worms; LM and VS did the doxycycline and motility experiments on-plate. MC, GK and LM did the data analysis and image processing. All authors contributed to write the manuscript. All authors read and approved the final manuscript.
About this article
- Caenorhabditis elegans
- Neurodegenerative disease
- Amyotrophic lateral sclerosis (ALS)
- Huntington disease (HD)
- Doxycycline treatment
- Protein aggregation
- Longitudinal time-resolved analysis
- High-resolution imaging
- Worm immobilization
- Temperature control